@article {Fichtinger2008b, title = {Robotic assistance for ultrasound-guided prostate brachytherapy}, journal = {Medical Image Analysis}, volume = {12}, number = {5}, year = {2008}, month = {06/2008}, pages = {535}, publisher = {Queen{\textquoteright}s University; 25 Union Street;725 Goodwin Hall; Kingston; ON; Canada gabor@cs queensu ca}, abstract = {

We present a robotically assisted prostate brachytherapy system, test results in training phantoms, Phase-I clinical trials The system consists of a transrectal ultrasound (TRUS), a spatially co-registered robot,fully integrated with an FDA-approved commercial treatment planning system The salient feature of the system is a small parallel robot affixed to the mounting posts of the template The robot replaces the template interchangeably,using the same coordinate system Established clinical hardware,workflow, calibration remain intact In all phantom experiments,we recorded the first insertion attempt without adjustment All clinically relevant locations in the prostate were reached Non-parallel needle trajectories were achieved The pre-insertion transverse, rotational errors (measured with a Polaris optical tracker relative to the template{\textquoteright}s coordinate frame) were 0 25 mm (STD=0 17 mm), 0 75 degrees (STD=0 37 degrees) In phantoms,needle tip placement errors measured in TRUS were 1 04 mm (STD=0 50mm) A Phase-I clinical feasibility, safety trial has been successfully completed with the system We encountered needle tip positioning errors of a magnitude greater than 4mm in only 2 of 179 robotically guided needles,in contrast to manual template guidance where errors of this magnitude are much more common Further clinical trials are necessary to determine whether the apparent benefits of the robotic assistant will lead to improvements in clinical efficacy, outcomes

}, keywords = {Brachytherapy, Computer-Assisted, Equipment Design, Equipment Failure Analysis, Humans, instrumentation, instrumentation/methods, Interventional, Male, Prostatic Neoplasms, Radiotherapy, radiotherapy/ultrasonography, Robotics, Ultrasonography}, doi = {10 1016/j media 2008 06 002}, url = {http://dx doi org/10 1016/j media 2008 06 002}, author = {Gabor Fichtinger and Jonathan Fiene and Christopher Kennedy and Gernot Kronreif and Iulian Iordachita and Danny Y Song and E. Clif Burdette and Peter Kazanzides} } @article {Tokuda2008, title = {Software strategy for robotic transperineal prostate therapy in closed-bore MRI}, journal = {Medical image computing and computer-assisted intervention (MICCAI)}, volume = {11}, number = {Pt 2}, year = {2008}, pages = {701{\textendash}709}, publisher = {Department of Radiology, Brigham, Women{\textquoteright}s Hospital, Harvard Medical School, 75 Francis St , Boston, MA 02115, USA tokuda@bwh harvard edu}, abstract = {

A software strategy to provide intuitive navigation for MRI-guided robotic transperineal prostate therapy is presented In the system, the robot control unit, the MRI scanner, and open-source navigation software are connected to one another via Ethernet to exchange commands, coordinates, and images Six states of the system called "workphases" are defined based on the clinical scenario to synchronize behaviors of all components The wizard-style user interface allows easy following of the clinical workflow On top of this framework, the software provides features for intuitive needle guidance: interactive target planning; 3D image visualization with current needle position; treatment monitoring through real-time MRI These features are supported by calibration of robot, image coordinates by the fiducial-based registration The performance test shows that the registration error of the system was 2 6 mm in the prostate area, and it displayed real-time 2D image 1 7 s after the completion of image acquisition

}, keywords = {Brachytherapy, Computer-Assisted, Humans, Image Interpretation, Interventional, Magnetic Resonance Imaging, Male, methods, Needles, pathology/radiotherapy, Prostatic Neoplasms, Radiotherapy, Robotics, Software, Software Design}, author = {Junichi Tokuda and Gregory Fischer and Csaba Csoma and Simon DiMaio and David Gobbi and Gabor Fichtinger and Clare M Tempany and Nobuhiko Hata} } @article {DiMaio2007b, title = {Challenges in image-guided therapy system design}, journal = {Neuroimage}, volume = {37 Suppl 1}, year = {2007}, pages = {S144{\textendash}S151}, publisher = {Brigham, Women{\textquoteright}s Hospital, 75 Francis St , Boston, MA 02115, USA simond@bwh harvard edu}, abstract = {

System development for image-guided therapy (IGT), or image-guided interventions (IGI), continues to be an area of active interest across academic, industry groups This is an emerging field that is growing rapidly: major academic institutions, medical device manufacturers have produced IGT technologies that are in routine clinical use, dozens of high-impact publications are published in well regarded journals each year, and several small companies have successfully commercialized sophisticated IGT systems In meetings between IGT investigators over the last two years, a consensus has emerged that several key areas must be addressed collaboratively by the community to reach the next level of impact, efficiency in IGT research, development to improve patient care These meetings culminated in a two-day workshop that brought together several academic, industrial leaders in the field today The goals of the workshop were to identify gaps in the engineering infrastructure available to IGT researchers, develop the role of research funding agencies, the recently established US-based National Center for Image Guided Therapy (NCIGT), and ultimately to facilitate the transfer of technology among research centers that are sponsored by the National Institutes of Health (NIH) Workshop discussions spanned many of the current challenges in the development, deployment of new IGT systems Key challenges were identified in a number of areas, including: validation standards; workflows, use-cases, and application requirements; component reusability;, device interface standards This report elaborates on these key points, proposes research challenges that are to be addressed by a joint effort between academic, industry, and NIH participants

}, keywords = {adverse effects/instrumentation/trends, Algorithms, Computational Biology, Computer Systems, Computer-Assisted, Humans, instrumentation/trends, Knowledge Bases, Models, National Institutes of Health (U S ), Neurosurgical Procedures, Organizational, Outcome Assessment (Health Care), Reproducibility of Results, Robotics, Software, Surgery, Treatment Outcome, United States}, doi = {10 1016/j neuroimage 2007 04 026}, url = {http://dx doi org/10 1016/j neuroimage 2007 04 026}, author = {Simon DiMaio and Tina Kapur and Kevin Cleary and Stephen Aylward and Peter Kazanzides and Kirby Vosburgh and Randy Ellis and James Duncan and Keyvan Farahani and Heinz Lemke and T. M. Peters and William Bill Lorensen and David Gobbi and John Haller and Laurence Larry Clarke and Stephen Pizer and Russell H. Taylor and Robert L. Galloway, Jr. and Gabor Fichtinger and Nobuhiko Hata and Kimberly Lawson and Clare M Tempany and Ron Kikinis and Ferenc Jolesz} } @article {Krieger2007a, title = {Design and preliminary accuracy studies of an MRI-guided transrectal prostate intervention system}, journal = {Medical image computing and computer-assisted intervention (MICCAI)}, volume = {10}, number = {Pt 2}, year = {2007}, pages = {59{\textendash}67}, publisher = {Department of Mechanical Engineering, Johns Hopkins University, Baltimore, USA}, abstract = {

This paper reports a novel system for magnetic resonance imaging (MRI) guided transrectal prostate interventions, such as needle biopsy, fiducial marker placement, and therapy delivery The system utilizes a hybrid tracking method, comprised of passive fiducial tracking for initial registration, subsequent incremental motion measurement along the degrees of freedom using fiber-optical encoders, mechanical scales Targeting accuracy of the system is evaluated in prostate phantom experiments Achieved targeting accuracy, procedure times were found to compare favorably with existing systems using passive, active tracking methods Moreover, the portable design of the system using only standard MRI image sequences, minimal custom scanner interfacing allows the system to be easily used on different MRI scanners

}, keywords = {Algorithms, Artificial Intelligence, Automated, Biopsy, Computer-Assisted, Equipment Design, Equipment Failure Analysis, Humans, Image Enhancement, Image Interpretation, Imaging, instrumentation/methods, Interventional, Magnetic Resonance Imaging, Male, methods, Needle, pathology/surgery, Pattern Recognition, Pilot Projects, Prostatectomy, Prostatic Neoplasms, Rectum, Reproducibility of Results, Robotics, Sensitivity, Specificity, Three-Dimensional, User-Computer Interface}, author = {Axel Krieger and Csaba Csoma and Iulian Iordachita and Peter Guion and Anurag K. Singh and Gabor Fichtinger and Whitcomb, Louis L.} } @article {Kazanzides2007, title = {Development of an image-guided robot for small animal research}, journal = {Journal of Computer Aided Surgery}, volume = {12}, number = {6}, year = {2007}, month = {Nov}, pages = {357{\textendash}365}, publisher = {Department of Computer Science, The Johns Hopkins University, Baltimore, Maryland 21218, USA pkaz@jhu edu}, abstract = {

We developed a robot system that can be used for image-guided experimental procedures on small animals, where the goal is to perform physical actions at specific positions identified on a preoperative image The animal is first placed in a fixture that is compatible with all imaging systems of interest, including PET, SPECT, CT, MRI After imaging, the fixture is attached, registered to the robot system, where the image-guided intervention is performed This system has been applied to perform pO(2) measurements with physical probes based on tumor hypoxia images obtained in an animal PET scanner This paper focuses on the design, validation of the robot system The validation is performed using a phantom, includes a new method for estimating the Fiducial Localization Error (FLE) that is based on the measured Fiducial Distance Error (FDE) The results indicate that the robot system can position the measurement probe at a defined target with a mean error that is less than 0 4 mm

}, keywords = {Algorithms, analysis, Animals, Imaging, Laboratory, methods, Oxygen, Partial Pressure, Phantoms, Positron-Emission Tomography, Robotics}, doi = {10 1080/10929080701732538}, url = {http://dx doi org/10 1080/10929080701732538}, author = {Peter Kazanzides and Jenghwa Chang and Iulian Iordachita and Jack Li and Clifton Ling and Gabor Fichtinger} } @article {DiMaio2007, title = {Dynamic MRI scan plane control for passive tracking of instruments and devices}, journal = {Medical image computing and computer-assisted intervention (MICCAI)}, volume = {10}, number = {Pt 2}, year = {2007}, pages = {50{\textendash}58}, publisher = {Brigham, Women{\textquoteright}s Hospital, Harvard Medical School, Boston, MA, USA}, abstract = {

This paper describes a novel image-based method for tracking robotic mechanisms, interventional devices during Magnetic Resonance Image (MRI)-guided procedures It takes advantage of the multi-planar imaging capabilities of MRI to optimally image a set of localizing fiducials for passive motion tracking in the image coordinate frame The imaging system is servoed to adaptively position the scan plane based on automatic detection, localization of fiducial artifacts directly from the acquired image stream This closed-loop control system has been implemented using an open-source software framework, currently operates with GE MRI scanners Accuracy, performance were evaluated in experiments, the results of which are presented here

}, keywords = {Algorithms, Anatomy, Artificial Intelligence, Automated, Computer-Assisted, Cross-Sectional, Humans, Image Enhancement, Image Interpretation, Imaging, instrumentation/methods, Magnetic Resonance Imaging, methods, Pattern Recognition, Phantoms, Reproducibility of Results, Robotics, Sensitivity, Specificity, Surgery, Three-Dimensional, User-Computer Interface}, author = {Simon DiMaio and Eigil Samset and Gregory Fischer and Iulian Iordachita and Gabor Fichtinger and Ferenc Jolesz and Clare M Tempany} } @article {Krupa2007, title = {Real-time tissue tracking with B-mode ultrasound using speckle and visual servoing}, journal = {Tenth International Conference on Medical Image Computing and Computer-Assisted Intervention (MICCAI)}, volume = {10}, number = {Pt 2}, year = {2007}, pages = {1{\textendash}8}, publisher = {IRISA - INRIA Rennes, France alexandre krupa@irisa fr}, abstract = {

We present a method for real-time tracking of moving soft tissue with B-mode ultrasound (US) The method makes use of the speckle information contained in the US images to estimate the in-plane, out-of-plane motion of a fixed target relative to the ultrasound scan plane The motion information is then used as closed-loop feedback to a robot which corrects for the target motion The concept is demonstrated for translation motions in an experimental setup consisting of an ultrasound speckle phantom, a robot for simulating tissue motion, and a robot that performs motion stabilization from US images This concept shows promise for US-guided procedures that require real-time motion tracking, compensation

}, keywords = {Algorithms, Artificial Intelligence, Computer Systems, Computer-Assisted, Equipment Design, Equipment Failure Analysis, Image Enhancement, Image Interpretation, Imaging, instrumentation/methods, Interventional, methods, Pilot Projects, Reproducibility of Results, Robotics, Sensitivity, Specificity, Surgery, Three-Dimensional, Ultrasonography, User-Computer Interface}, author = {Alexandre Krupa and Gabor Fichtinger and Gregory D. Hager} } @article {DiMaio2007a, title = {Robot-assisted needle placement in open MRI: system architecture, integration and validation}, journal = {Journal of Computer Aided Surgery}, volume = {12}, number = {1}, year = {2007}, month = {Jan}, pages = {15{\textendash}24}, publisher = {Brigham, Women{\textquoteright}s Hospital, Harvard Medical School, Boston, Massachusetts 02115, USA simond@bwh harvard edu}, abstract = {

In prostate cancer treatment, there is a move toward targeted interventions for biopsy, therapy, which has precipitated the need for precise image-guided methods for needle placement This paper describes an integrated system for planning, performing percutaneous procedures with robotic assistance under MRI guidance A graphical planning interface allows the physician to specify the set of desired needle trajectories, based on anatomical structures, lesions observed in the patient{\textquoteright}s registered pre-operative, pre-procedural MR images, immediately prior to the intervention in an open-bore MRI scanner All image-space coordinates are automatically computed, and are used to position a needle guide by means of an MRI-compatible robotic manipulator, thus avoiding the limitations of the traditional fixed needle template Automatic alignment of real-time intra-operative images aids visualization of the needle as it is manually inserted through the guide Results from in-scanner phantom experiments are provided

}, keywords = {Biopsy, diagnosis, Humans, instrumentation/methods, Magnetic Resonance Imaging, Male, Needle, Neuronavigation, Prostatic Neoplasms, Robotics}, doi = {10 1080/10929080601168254}, url = {http://dx doi org/10 1080/10929080601168254}, author = {Simon DiMaio and Steve Pieper and Kiyoyuki Chinzei and Nobuhiko Hata and Steven Haker and Daniel Kacher and Gabor Fichtinger and Clare M Tempany and Ron Kikinis} } @article {Fichtinger2007c, title = {Robotic assistance for ultrasound guided prostate brachytherapy}, journal = {Medical Image Computing and Computer-Assisted Intervention (MICCAI)}, volume = {4791}, year = {2007}, pages = {119{\textendash}127}, publisher = {The Johns Hopkins University, Baltimore, Maryland, USA gabor@cs jhu edu}, abstract = {

We present a robotically assisted prostate brachytherapy system, test results in training phantoms The system consists of a transrectal ultrasound (TRUS), a spatially co-registered robot integrated with an FDA-approved commercial treatment planning system The salient feature of the system is a small parallel robot affixed to the mounting posts of the template The robot replaces the template interchangeably, uses the same coordinate system Established clinical hardware, workflow, calibration are left intact In these experiments, we recorded the first insertion attempt without adjustment All clinically relevant locations were reached Non-parallel needle trajectories were achieved The pre-insertion transverse, rotational errors (measured with Polaris optical tracker relative to the template{\textquoteright}s coordinate frame) were 0 25mm (STD = 0 17mm), 0 75 degrees (STD = 0 37 degrees) The needle tip placement errors measured in TRUS were 1 04mm (STD = 0 50mm) The system is in Phase-I clinical feasibility, safety trials, under Institutional Review Board approval

}, keywords = {Brachytherapy, Computer-Assisted, Equipment Design, Equipment Failure Analysis, Humans, instrumentation/methods, Interventional, Male, Prostatic Neoplasms, Radiotherapy, Radiotherapy Planning, radiotherapy/ultrasonography, Robotics, Ultrasonography}, author = {Gabor Fichtinger and Jonathan Fiene and Christopher Kennedy and Gernot Kronreif and Iulian Iordachita and Danny Y Song and E. Clif Burdette and Peter Kazanzides} } @article {Fischer2007b, title = {Robotic assistant for transperineal prostate interventions in 3T closed MRI}, journal = {Medical image computing and computer-assisted intervention (MICCAI)}, volume = {10}, number = {Pt 1}, year = {2007}, pages = {425{\textendash}433}, publisher = {Center for Computer Integrated Surgery, Johns Hopkins University, USA gfischer@jhu edu}, abstract = {

Numerous studies have demonstrated the efficacy of image-guided needle-based therapy, biopsy in the management of prostate cancer The accuracy of traditional prostate interventions performed using transrectal ultrasound (TRUS) is limited by image fidelity, needle template guides, needle deflection, tissue deformation Magnetic Resonance Imaging (MRI) is an ideal modality for guiding, monitoring such interventions due to its excellent visualization of the prostate, its sub-structure, surrounding tissues We have designed a comprehensive robotic assistant system that allows prostate biopsy, brachytherapy procedures to be performed entirely inside a 3T closed MRI scanner We present a detailed design of the robotic manipulator, an evaluation of its usability, MR compatibility

}, keywords = {anatomy /\&/ histology/surgery, Computer-Assisted, Equipment Design, Equipment Failure Analysis, Image Interpretation, instrumentation/methods, Interventional, Magnetic Resonance Imaging, Perineum, Prostatectomy, Robotics, Surgery}, author = {Gregory Fischer and Simon DiMaio and Iulian Iordachita and Gabor Fichtinger} } @article {Kazanzides2006, title = {Design, validation of an image-guided robot for small animal research}, journal = {Medical image computing and computer-assisted intervention (MICCAI)}, volume = {9}, number = {Pt 1}, year = {2006}, pages = {50{\textendash}57}, publisher = {Department of Computer Science, Johns Hopkins University, USA pkaz@jhu edu}, abstract = {

We developed an image-guided robot system to achieve highly accurate placement of thin needles, probes into in-vivo rodent tumor tissue in a predefined pattern that is specified on a preoperative image This system can be used for many experimental procedures where the goal is to correlate a set of physical measurements with a corresponding set of image intensities or, more generally, to perform a physical action at a set of anatomic points identified on a preoperative image This paper focuses on the design, validation of the robot system, where the first application is to insert oxygen measurement probes in a three-dimensional (3D) grid pattern defined with respect to a PET scan of a tumor The design is compatible with CT, MRI, which we plan to use to identify targets for biopsy, for the injection of adenoviral sequences for gene therapy The validation is performed using a phantom, includes a new method for estimating the Fiducial Localization Error (FLE) based on the measured Fiducial Distance Error (FDE)

}, keywords = {Animals, Biopsy, Computer-Assisted, Equipment Design, Equipment Failure Analysis, Imaging, Injections, instrumentation/methods, instrumentation/methods/veterinary, Needle, Phantoms, Positron-Emission Tomography, Research, Robotics, Surgery}, author = {Peter Kazanzides and Jenghwa Chang and Iulian Iordachita and Jack Li and Clifton Ling and Gabor Fichtinger} } @article {DiMaio2006b, title = {Robot-assisted needle placement in open-MRI: system architecture, integration and validation}, journal = {Studies in health technology and informatics}, volume = {119}, year = {2006}, pages = {126{\textendash}131}, publisher = {Brigham, Women{\textquoteright}s Hospital, Harvard Medical School, USA simond@bwh harvard edu}, abstract = {

This work describes an integrated system for planning, performing percutaneous procedures-such as prostate biopsy-with robotic assistance under MRI-guidance The physician interacts with a planning interface in order to specify the set of desired needle trajectories, based on anatomical structures, lesions observed in the patient{\textquoteright}s MR images All image-space coordinates are automatically computed, and used to position a needle guide by means of an MRI-compatible robotic manipulator, thus avoiding the limitations of the traditional fixed needle template Direct control of real-time imaging aids visualization of the needle as it is manually inserted through the guide Results from in-scanner phantom experiments are provided

}, keywords = {Biopsy, Computer Systems, diagnosis, Humans, instrumentation, Magnetic Resonance Imaging, Male, Needle, Prostatic Neoplasms, Robotics}, author = {Simon DiMaio and Steve Pieper and Kiyoyuki Chinzei and Nobuhiko Hata and Emese Balogh and Gabor Fichtinger and Clare M Tempany and Ron Kikinis} } @article {Fichtinger2006, title = {Robotically assisted prostate brachytherapy with transrectal ultrasound guidance{\textendash}Phantom experiments}, journal = {Brachytherapy}, volume = {5}, number = {1}, year = {2006}, pages = {14{\textendash}26}, publisher = {Engineering Research Center, The Johns Hopkins University, Baltimore, MD 21218-2682, USA gabor@cs jhu edu}, abstract = {

PURPOSE: To report the preliminary experimental results obtained with a robot-assisted transrectal ultrasound (TRUS)-guided prostate brachytherapy system METHODS AND MATERIALS: The system consists of a TRUS unit, a spatially coregistered needle insertion robot, and an FDA-approved treatment planning, image-registered implant system The robot receives each entry/target coordinate pair of the implant plan, inserts a preloaded needle, and then the seeds are deposited The needles/sources are tracked in TRUS, thus allowing the plan to be updated as the procedure progresses RESULTS: The first insertion attempt was recorded for each needle, without adjustment All clinically relevant locations were reached in a prostate phantom Nonparallel, parallel needle trajectories were demonstrated Based on TRUS, the average transverse placement error was 2 mm (worst case 2 5 mm, 80\% less than 2 mm), and the average sagittal error was 2 5 mm (worst case 5 0 mm, 70\% less than 2 5 mm) CONCLUSIONS: The concept, technical viability of robot-assisted brachytherapy were demonstrated in phantoms The kinematically decoupled robotic assistant device is inherently safe Overall performance was promising, but further optimization is necessary to prove the possibility of improved dosimetry

}, keywords = {Anatomic, Brachytherapy, Endosonography, Equipment Design, Humans, instrumentation, Male, methods, Models, Prostatic Neoplasms, radiotherapy/ultrasonography, Rectum, Robotics}, doi = {10 1016/j brachy 2005 10 003}, url = {http://dx doi org/10 1016/j brachy 2005 10 003}, author = {Gabor Fichtinger and E. Clif Burdette and Attila Tanacs and Alexandru Patriciu and Dumitru Mazilu and Whitcomb, Louis L. and Dan Stoianovici} } @article {Krieger2005a, title = {Design of a novel MRI compatible manipulator for image guided prostate interventions}, journal = {IEEE Transactions on Biomedical Engineering}, volume = {52}, number = {2}, year = {2005}, month = {Feb}, pages = {306{\textendash}313}, abstract = {

This paper reports a novel remotely actuated manipulator for access to prostate tissue under magnetic resonance imaging guidance (APT-MRI) device, designed for use in a standard high-field MRI scanner The device provides three-dimensional MRI guided needle placement with millimeter accuracy under physician control Procedures enabled by this device include MRI guided needle biopsy, fiducial marker placements, and therapy delivery Its compact size allows for use in both standard cylindrical, open configuration MRI scanners Preliminary in vivo canine experiments, first clinical trials are reported

}, keywords = {Animals, Biopsy, Computer-Assisted, Dogs, Equipment Design, Equipment Failure Analysis, Feasibility Studies, Imaging, instrumentation/methods, Magnetic Resonance Imaging, Male, Micromanipulation, Needle, pathology/surgery, Prostate, Reproducibility of Results, Robotics, Sensitivity, Specificity, Surgery, Three-Dimensional}, doi = {10 1109/TBME 2004 840497}, url = {http://dx doi org/10 1109/TBME 2004 840497}, author = {Axel Krieger and Robert Susil and Cynthia Menard and Jonathan Coleman and Gabor Fichtinger and Ergin Atalar and Whitcomb, Louis L.} } @article {Cleary2005, title = {Precision placement of instruments for minimally invasive procedures using a "needle driver" robot}, journal = {international journal of medical robotics computer assisted surgery MRCAS}, volume = {1}, number = {2}, year = {2005}, month = {Jan}, pages = {40{\textendash}47}, abstract = {

Medical practice continues to move toward less invasive procedures Many of these procedures require the precision placement of a needle in the anatomy Over the past several years, our research team has been investigating the use of a robotic needle driver to assist the physician in this task This paper summarizes our work in this area The robotic system is briefly described, followed by a description of a clinical trial in spinal nerve blockade The robot was used under joystick control to place a 22 gauge needle in the spines of 10 patients using fluoroscopic imaging The results were equivalent to the current manual procedure We next describe our follow-up clinical application in lung biopsy for lung cancer screening under CT fluoroscopy The system concept is discussed, the results of a phantom study are presented A start-up company named ImageGuide has recently been formed to commercialize the robot Their revised robot design is presented, along with plans to install a ceiling-mounted version of the robot in the CT fluoroscopy suite at Georgetown University

}, keywords = {Biopsy, Cadaver, Equipment Design, Fluoroscopy, Humans, Imaging, instrumentation, instrumentation/methods, Lung, Lung Neoplasms, methods, Minimally Invasive, Needles, Nerve Block, pathology, pathology/radiography, Phantoms, Randomized Controlled Trials as Topic, Robotics, Spinal Nerves, Surgical Procedures, Tomography, X-Ray Computed}, doi = {10 1002/rcs 40}, url = {http://dx doi org/10 1002/rcs 40}, author = {Kevin Cleary and Vance Watson and David Lindisch and Russell H. Taylor and Gabor Fichtinger and Sheng Xu and Charles White and John Donlon and Michael Taylor and Alexandru Patriciu and Dumitru Mazilu and Dan Stoianovici} } @article {Cleary2004, title = {Medical Robotics workshop{\textendash}MRWS}, journal = {Journal of Computer Aided Surgery}, volume = {9}, number = {4}, year = {2004}, pages = {167{\textendash}171}, keywords = {Biomedical Research, Computer-Assisted, Humans, Operative, Robotics, Surgery, Surgical Procedures}, doi = {10 1080/10929080500097695}, url = {http://dx doi org/10 1080/10929080500097695}, author = {Kevin Cleary and Brian Davies and Gabor Fichtinger and Jocelyne Troccaz and Tim Lueth and Vance Watson} } @article {Boctor2003b, title = {Virtual remote center of motion control for needle placement robots}, journal = {Journal of Computer Aided Surgery}, volume = {9}, number = {5}, year = {2003}, pages = {175{\textendash}183}, abstract = {

OBJECTIVE: We present an algorithm that enables percutaneous needle-placement procedures to be performed with unencoded, unregistered, minimally calibrated robots while removing the constraint of placing the needle tip on a mechanically enforced Remote Center of Motion (RCM) MATERIALS AND METHODS: The algorithm requires only online tracking of the surgical tool, a five-degree-of-freedom (5-DOF) robot comprising three prismatic DOF, two rotational DOF An incremental adaptive motion control cycle guides the needle to the insertion point, also orients it to align with the target-entry-point line The robot executes RCM motion without having a physically constrained fulcrum point RESULTS: The proof-of-concept prototype system achieved 0 78 mm translation accuracy, 1 4 degrees rotational accuracy (this is within the tracker accuracy) within 17 iterative steps (0 5-1 s) CONCLUSION: This research enables robotic assistant systems for image-guided percutaneous procedures to be prototyped/constructed more quickly, less expensively than has been previously possible Since the clinical utility of such systems is clear, has been demonstrated in the literature, our work may help promote widespread clinical adoption of this technology by lowering system cost, complexity

}, keywords = {Algorithms, Computer-Assisted, Equipment Design, Imaging, instrumentation, Needles, Robotics, Surgery, Three-Dimensional}, doi = {10 1080/10929080500097661}, url = {http://dx doi org/10 1080/10929080500097661}, author = {Emad Boctor and Robert Webster and Herve Mathieu and Allison Okamura and Gabor Fichtinger} } @article {Masamune2001, title = {System for robotically assisted percutaneous procedures with computed tomography guidance}, journal = {Journal of Computer Aided Surgery}, volume = {6}, number = {6}, year = {2001}, pages = {370{\textendash}383}, abstract = {

We present the prototype of an image-guided robotic system for accurate, consistent placement of percutaneous needles in soft-tissue targets under CT guidance inside the gantry of a CT scanner The couch-mounted system consists of a seven-degrees-of-freedom passive mounting arm, a remote center-of-motion robot, and a motorized needle-insertion device Single-image-based coregistration of the robot, image space is achieved by stereotactic localization using a miniature version of the BRW head frame built into the radiolucent needle driver The surgeon plans, controls the intervention in the scanner room on a desktop computer that receives DICOM images from the scanner The system does not need calibration, employs pure image-based registration, and does not utilize any vendor-specific hardware or software features In the open air, where there is no needle-tissue interaction, we systematically achieved an accuracy better than 1 mm in hitting targets at 5-8 cm from the fulcrum point In the phantom, the orientation accuracy was 0 6 degrees, and the distance between the needle tip, the target was 1 04 mm Experiments indicated that this robotic system is suitable for a variety of percutaneous clinical applications

}, keywords = {Biopsy, Computer-Assisted, Humans, Image Processing, Imaging, instrumentation, instrumentation/methods, Needle, Phantoms, Robotics, Surgery, Tomography, User-Computer Interface, X-Ray Computed}, doi = {10 1002/igs 10024}, url = {http://dx doi org/10 1002/igs 10024}, author = {Ken Masamune and Gabor Fichtinger and Alexandru Patriciu and Robert Susil and Russell H. Taylor and Louis Kavoussi and James Anderson and Ichiro Sakuma and Takeyoshi Dohi and Dan Stoianovici} }