@article {Fichtinger2005, title = {Image overlay for CT-guided needle insertions}, journal = {Journal of Computer Aided Surgery}, volume = {10}, number = {4}, year = {2005}, month = {Jul}, pages = {241{\textendash}255}, abstract = {

OBJECTIVE: We present a 2D image overlay device to assist needle placement on computed tomography (CT) scanners MATERIALS AND METHODS: The system consists of a flat display, a semitransparent mirror mounted on the gantry When the physician looks at the patient through the mirror, the CT image appears to be floating inside the body with correct size, position as if the physician had 2D {\textquoteright}X-ray vision{\textquoteright} The physician draws the optimal path on the CT image The composite image is rendered on the display, thus reflected in the mirror The reflected image is used to guide the physician in the procedure In this article, we describe the design, various embodiments of the 2D image overlay system, followed by the results of phantom, cadaver experiments in multiple clinical applications RESULTS: Multiple skeletal targets were successfully accessed with one insertion attempt Generally, successful access was recorded on liver targets when a clear path opened, but the number of attempts, accuracy showed variability because of occasional lack of access Soft tissue deformation further reduced the accuracy, consistency in comparison to skeletal targets CONCLUSION: The system demonstrated strong potential for reducing faulty needle insertion attempts, thereby reducing X-ray dose, patient discomfort

}, keywords = {Animals, Biopsy, Cadaver, Calibration, Computer-Assisted, Equipment Design, Humans, Image Processing, Imaging, instrumentation, Interventional, Needle, Phantoms, Radiography, Swine, Tomography, User-Computer Interface, X-Ray Computed}, doi = {10 1080/10929080500230486}, url = {http://dx doi org/10 1080/10929080500230486}, author = {Gabor Fichtinger and Anton Degeut and Gregory Fischer and Iulian Iordachita and Emese Balogh and Ken Masamune and Russell H. Taylor and Laura Fayad and Michelle de Oliveira and James Zinreich} } @article {Shen2004a, title = {Optimized prostate biopsy via a statistical atlas of cancer spatial distribution}, journal = {Medical image analysis}, volume = {8}, number = {2}, year = {2004}, month = {Jun}, pages = {139{\textendash}150}, abstract = {

A methodology is presented for constructing a statistical atlas of spatial distribution of prostate cancer from a large patient cohort, and it is used for optimizing needle biopsy An adaptive-focus deformable model is used for the spatial normalization, registration of 100 prostate histological samples, which were provided by the Center for Prostate Disease Research of the US Department of Defense, resulting in a statistical atlas of spatial distribution of prostate cancer Based on this atlas, a statistical predictive model was developed to optimize the needle biopsy sites, by maximizing the probability of detecting cancer Experimental results using cross-validation show that the proposed method can detect cancer with a 99\% success rate using seven needles, in these samples

}, keywords = {Algorithms, Biological, Biopsy, Cohort Studies, Computer Simulation, Computer-Assisted, Forecasting, Humans, Image Processing, Imaging, Male, Models, Needle, pathology, Probability, Prostate, Prostatic Neoplasms, Reproducibility of Results, Statistical, statistics /\&/ numerical data, Three-Dimensional}, doi = {10 1016/j media 2003 11 002}, url = {http://dx doi org/10 1016/j media 2003 11 002}, author = {Dinggang Shen and Zhiqiang Lao and Jianchao Zeng and Wei Zhang and Isabel Sesterhenn and Leon Sun and Judd Moul and Edward Herskovits and Gabor Fichtinger and Christos Davatzikos} } @article {Masamune2001, title = {System for robotically assisted percutaneous procedures with computed tomography guidance}, journal = {Journal of Computer Aided Surgery}, volume = {6}, number = {6}, year = {2001}, pages = {370{\textendash}383}, abstract = {

We present the prototype of an image-guided robotic system for accurate, consistent placement of percutaneous needles in soft-tissue targets under CT guidance inside the gantry of a CT scanner The couch-mounted system consists of a seven-degrees-of-freedom passive mounting arm, a remote center-of-motion robot, and a motorized needle-insertion device Single-image-based coregistration of the robot, image space is achieved by stereotactic localization using a miniature version of the BRW head frame built into the radiolucent needle driver The surgeon plans, controls the intervention in the scanner room on a desktop computer that receives DICOM images from the scanner The system does not need calibration, employs pure image-based registration, and does not utilize any vendor-specific hardware or software features In the open air, where there is no needle-tissue interaction, we systematically achieved an accuracy better than 1 mm in hitting targets at 5-8 cm from the fulcrum point In the phantom, the orientation accuracy was 0 6 degrees, and the distance between the needle tip, the target was 1 04 mm Experiments indicated that this robotic system is suitable for a variety of percutaneous clinical applications

}, keywords = {Biopsy, Computer-Assisted, Humans, Image Processing, Imaging, instrumentation, instrumentation/methods, Needle, Phantoms, Robotics, Surgery, Tomography, User-Computer Interface, X-Ray Computed}, doi = {10 1002/igs 10024}, url = {http://dx doi org/10 1002/igs 10024}, author = {Ken Masamune and Gabor Fichtinger and Alexandru Patriciu and Robert Susil and Russell H. Taylor and Louis Kavoussi and James Anderson and Ichiro Sakuma and Takeyoshi Dohi and Dan Stoianovici} }