@article {Fichtinger2008b, title = {Robotic assistance for ultrasound-guided prostate brachytherapy}, journal = {Medical Image Analysis}, volume = {12}, number = {5}, year = {2008}, month = {06/2008}, pages = {535}, publisher = {Queen{\textquoteright}s University; 25 Union Street;725 Goodwin Hall; Kingston; ON; Canada gabor@cs queensu ca}, abstract = {

We present a robotically assisted prostate brachytherapy system, test results in training phantoms, Phase-I clinical trials The system consists of a transrectal ultrasound (TRUS), a spatially co-registered robot,fully integrated with an FDA-approved commercial treatment planning system The salient feature of the system is a small parallel robot affixed to the mounting posts of the template The robot replaces the template interchangeably,using the same coordinate system Established clinical hardware,workflow, calibration remain intact In all phantom experiments,we recorded the first insertion attempt without adjustment All clinically relevant locations in the prostate were reached Non-parallel needle trajectories were achieved The pre-insertion transverse, rotational errors (measured with a Polaris optical tracker relative to the template{\textquoteright}s coordinate frame) were 0 25 mm (STD=0 17 mm), 0 75 degrees (STD=0 37 degrees) In phantoms,needle tip placement errors measured in TRUS were 1 04 mm (STD=0 50mm) A Phase-I clinical feasibility, safety trial has been successfully completed with the system We encountered needle tip positioning errors of a magnitude greater than 4mm in only 2 of 179 robotically guided needles,in contrast to manual template guidance where errors of this magnitude are much more common Further clinical trials are necessary to determine whether the apparent benefits of the robotic assistant will lead to improvements in clinical efficacy, outcomes

}, keywords = {Brachytherapy, Computer-Assisted, Equipment Design, Equipment Failure Analysis, Humans, instrumentation, instrumentation/methods, Interventional, Male, Prostatic Neoplasms, Radiotherapy, radiotherapy/ultrasonography, Robotics, Ultrasonography}, doi = {10 1016/j media 2008 06 002}, url = {http://dx doi org/10 1016/j media 2008 06 002}, author = {Gabor Fichtinger and Jonathan Fiene and Christopher Kennedy and Gernot Kronreif and Iulian Iordachita and Danny Y Song and E. Clif Burdette and Peter Kazanzides} } @article {Fichtinger2007c, title = {Robotic assistance for ultrasound guided prostate brachytherapy}, journal = {Medical Image Computing and Computer-Assisted Intervention (MICCAI)}, volume = {4791}, year = {2007}, pages = {119{\textendash}127}, publisher = {The Johns Hopkins University, Baltimore, Maryland, USA gabor@cs jhu edu}, abstract = {

We present a robotically assisted prostate brachytherapy system, test results in training phantoms The system consists of a transrectal ultrasound (TRUS), a spatially co-registered robot integrated with an FDA-approved commercial treatment planning system The salient feature of the system is a small parallel robot affixed to the mounting posts of the template The robot replaces the template interchangeably, uses the same coordinate system Established clinical hardware, workflow, calibration are left intact In these experiments, we recorded the first insertion attempt without adjustment All clinically relevant locations were reached Non-parallel needle trajectories were achieved The pre-insertion transverse, rotational errors (measured with Polaris optical tracker relative to the template{\textquoteright}s coordinate frame) were 0 25mm (STD = 0 17mm), 0 75 degrees (STD = 0 37 degrees) The needle tip placement errors measured in TRUS were 1 04mm (STD = 0 50mm) The system is in Phase-I clinical feasibility, safety trials, under Institutional Review Board approval

}, keywords = {Brachytherapy, Computer-Assisted, Equipment Design, Equipment Failure Analysis, Humans, instrumentation/methods, Interventional, Male, Prostatic Neoplasms, Radiotherapy, Radiotherapy Planning, radiotherapy/ultrasonography, Robotics, Ultrasonography}, author = {Gabor Fichtinger and Jonathan Fiene and Christopher Kennedy and Gernot Kronreif and Iulian Iordachita and Danny Y Song and E. Clif Burdette and Peter Kazanzides} } @article {Kazanzides2006, title = {Design, validation of an image-guided robot for small animal research}, journal = {Medical image computing and computer-assisted intervention (MICCAI)}, volume = {9}, number = {Pt 1}, year = {2006}, pages = {50{\textendash}57}, publisher = {Department of Computer Science, Johns Hopkins University, USA pkaz@jhu edu}, abstract = {

We developed an image-guided robot system to achieve highly accurate placement of thin needles, probes into in-vivo rodent tumor tissue in a predefined pattern that is specified on a preoperative image This system can be used for many experimental procedures where the goal is to correlate a set of physical measurements with a corresponding set of image intensities or, more generally, to perform a physical action at a set of anatomic points identified on a preoperative image This paper focuses on the design, validation of the robot system, where the first application is to insert oxygen measurement probes in a three-dimensional (3D) grid pattern defined with respect to a PET scan of a tumor The design is compatible with CT, MRI, which we plan to use to identify targets for biopsy, for the injection of adenoviral sequences for gene therapy The validation is performed using a phantom, includes a new method for estimating the Fiducial Localization Error (FLE) based on the measured Fiducial Distance Error (FDE)

}, keywords = {Animals, Biopsy, Computer-Assisted, Equipment Design, Equipment Failure Analysis, Imaging, Injections, instrumentation/methods, instrumentation/methods/veterinary, Needle, Phantoms, Positron-Emission Tomography, Research, Robotics, Surgery}, author = {Peter Kazanzides and Jenghwa Chang and Iulian Iordachita and Jack Li and Clifton Ling and Gabor Fichtinger} }