@article {Sciubba2010,
	title = {Percutaneous computed tomography fluoroscopy-guided conformal ultrasonic ablation of vertebral tumors in a rabbit tumor model. Laboratory investigation.},
	journal = {J Neurosurg Spine},
	volume = {13},
	number = {6},
	year = {2010},
	month = {Dec},
	pages = {773{\textendash}779},
	publisher = {Department of Neurosurgery, Johns Hopkins University, Baltimore, Maryland 21287, USA. dsciubb1@jhmi.edu},
	abstract = {<p>Radiofrequency ablation (RFA) has proven to be effective for treatment of malignant and benign tumors in numerous anatomical sites outside the spine. The major challenge of using RFA for spinal tumors is difficulty protecting the spinal cord and nerves from damage. However, conforming ultrasound energy to match the exact anatomy of the tumor may provide successful ablation in such sensitive locations. In a rabbit model of vertebral body tumor, the authors have successfully ablated tumors using an acoustic ablator placed percutaneously via computed tomography fluoroscopic (CTF) guidance.Using CTF guidance, 12 adult male New Zealand White rabbits were injected with VX2 carcinoma cells in the lowest lumbar vertebral body. At 21 days, a bone biopsy needle was placed into the geographical center of the lesion, down which an acoustic ablator was inserted. Three multisensor thermocouple arrays were placed around the lesion to provide measurement of tissue temperature during ablation, at thermal doses ranging from 100 to 1,000,000 TEM (thermal equivalent minutes at 43{\textdegree}C), and tumor volumes were given a tumoricidal dose of acoustic energy. Animals were monitored for 24 hours and then sacrificed. Pathological specimens were obtained to determine the extent of tumor death and surrounding tissue damage. Measured temperature distributions were used to reconstruct volumetric doses of energy delivered to tumor tissue, and such data were correlated with pathological findings.All rabbits were successfully implanted with VX2 cells, leading to a grossly apparent spinal and paraspinal tissue mass. The CTF guidance provided accurate placement of the acoustic ablator in all tumors, as corroborated through gross and microscopic histology. Significant tumor death was noted in all specimens without collateral damage to nearby nerve tissue. Tissue destruction just beyond the margin of the tumor was noted in some but not all specimens. No neurological deficits occurred in response to ablation. Reconstruction of measured temperature data allowed accurate assessment of volumetric dose delivered to tissues.Using a rabbit intravertebral tumor model, the authors have successfully delivered tumoricidal doses of acoustic energy via a therapeutic ultrasound ablation probe placed percutaneously with CTF guidance. The authors have thus established the first technical and preclinical feasibility study of controlled ultrasound ablation of spinal tumors in vivo.</p>
},
	keywords = {Animals, Carcinoma, Catheter Ablation, Fluoroscopy, instrumentation/methods, Lumbar Vertebrae, Male, Neoplasm Transplantation, Rabbits, Spinal Neoplasms, therapy, Tomography, X-Ray Computed},
	doi = {10.3171/2010.5.SPINE09266},
	url = {http://dx.doi.org/10.3171/2010.5.SPINE09266},
	author = {Daniel M Sciubba and E. Clif Burdette and Jennifer J Cheng and William A Pennant and Joseph C Noggle and Rory J Petteys and Christopher Alix and Chris J Diederich and Gabor Fichtinger and Ziya L Gokaslan and Kieran P Murphy}
}
@article {Jain2007,
	title = {Intra-operative 3D guidance in prostate brachytherapy using a non-isocentric C-arm},
	journal = {Medical image computing and computer-assisted intervention (MICCAI)},
	volume = {10},
	number = {Pt 2},
	year = {2007},
	pages = {9{\textendash}17},
	publisher = {Johns Hopkins University, USA},
	abstract = {<p>Intra-operative guidance in Transrectal Ultrasound (TRUS) guided prostate brachytherapy requires localization of inserted radioactive seeds relative to the prostate Seeds were reconstructed using a typical C-arm, and exported to a commercial brachytherapy system for dosimetry analysis Technical obstacles for 3D reconstruction on a non-isocentric C-arm included pose-dependent C-arm calibration; distortion correction; pose estimation of C-arm images; seed reconstruction;, C-arm to TRUS registration In precision-machined hard phantoms with 40-100 seeds, we correctly reconstructed 99 8\% seeds with a mean 3D accuracy of 0 68 mm In soft tissue phantoms with 45-87 seeds, clinically realistic 15 degrees C-arm motion, we correctly reconstructed 100\% seeds with an accuracy of 1 3 mm The reconstructed 3D seed positions were then registered to the prostate segmented from TRUS In a Phase-1 clinical trial, so far on 4 patients with 66-84 seeds, we achieved intra-operative monitoring of seed distribution, dosimetry We optimized the 100\% prescribed iso-dose contour by inserting an average of 3 75 additional seeds, making intra-operative dosimetry possible on a typical C-arm, at negligible additional cost to the existing clinical installation</p>
},
	keywords = {Algorithms, Brachytherapy, Computer-Assisted, Equipment Design, Equipment Failure Analysis, Humans, Imaging, instrumentation/methods, Interventional, Intraoperative Care, Male, methods, Prostatic Neoplasms, Prosthesis Implantation, Radiographic Image Enhancement, Radiographic Image Interpretation, Radiography, radiography/radiotherapy, Radiotherapy, Reproducibility of Results, Sensitivity, Specificity, Three-Dimensional, Tomography, X-Ray Computed},
	author = {Ameet K. Jain and Anton Degeut and Iulian Iordachita and Gouthami Chintalapani and Jack Blevins and Le Yi and Elwood Armour and E. Clif Burdette and Danny Y Song and Gabor Fichtinger}
}
