@article {Foroughi2008, title = {Localization of pelvic anatomical coordinate system using US/atlas registration for total hip replacement}, journal = {Medical image computing and computer-assisted intervention (MICCAI)}, volume = {11}, number = {Pt 2}, year = {2008}, pages = {871{\textendash}879}, publisher = {Johns Hopkins University, Baltimore, MD, USA pezhman@cs jhu edu}, abstract = {

In Total Hip Replacement (THR) procedures, misalignment of the acetabular component can lead to dislocation, impingement For the successful alignment of acetabular component, precise estimation of pelvic anatomical coordinate system is necessary Conventional navigation systems use CT scan or fluoroscopy, or involve implanted bone fiducials or invasive probing of bony landmarks to locate the anatomical coordinate In this paper, an ultrasound-based approach is proposed that exploits prior knowledge about the anatomy of the pelvis in the form of a 3D surface atlas Tracked ultrasound images are utilized to extract sample points from the surface of the pelvis A generic coordinate system in the specific patient is localized by registering these points to a statistical atlas of the pelvis in which a canonical anatomical coordinate system had been defined This technique has been evaluated using simulation, dry bone, and cadaver experiments, was able to localize the anatomical coordinate system with the accuracy of about 1 degree

}, keywords = {Algorithms, Arthroplasty, Automated, Computer-Assisted, Hip, Hip Joint, Humans, Image Enhancement, Image Interpretation, Imaging, methods, Pattern Recognition, Pelvic Bones, Replacement, Reproducibility of Results, Sensitivity, Specificity, Subtraction Technique, Surgery, surgery/ultrasonography, Three-Dimensional, Ultrasonography}, author = {Pezhman Foroughi and Danny Y Song and Gouthami Chintalapani and Russell H. Taylor and Gabor Fichtinger} } @article {DiMaio2007b, title = {Challenges in image-guided therapy system design}, journal = {Neuroimage}, volume = {37 Suppl 1}, year = {2007}, pages = {S144{\textendash}S151}, publisher = {Brigham, Women{\textquoteright}s Hospital, 75 Francis St , Boston, MA 02115, USA simond@bwh harvard edu}, abstract = {

System development for image-guided therapy (IGT), or image-guided interventions (IGI), continues to be an area of active interest across academic, industry groups This is an emerging field that is growing rapidly: major academic institutions, medical device manufacturers have produced IGT technologies that are in routine clinical use, dozens of high-impact publications are published in well regarded journals each year, and several small companies have successfully commercialized sophisticated IGT systems In meetings between IGT investigators over the last two years, a consensus has emerged that several key areas must be addressed collaboratively by the community to reach the next level of impact, efficiency in IGT research, development to improve patient care These meetings culminated in a two-day workshop that brought together several academic, industrial leaders in the field today The goals of the workshop were to identify gaps in the engineering infrastructure available to IGT researchers, develop the role of research funding agencies, the recently established US-based National Center for Image Guided Therapy (NCIGT), and ultimately to facilitate the transfer of technology among research centers that are sponsored by the National Institutes of Health (NIH) Workshop discussions spanned many of the current challenges in the development, deployment of new IGT systems Key challenges were identified in a number of areas, including: validation standards; workflows, use-cases, and application requirements; component reusability;, device interface standards This report elaborates on these key points, proposes research challenges that are to be addressed by a joint effort between academic, industry, and NIH participants

}, keywords = {adverse effects/instrumentation/trends, Algorithms, Computational Biology, Computer Systems, Computer-Assisted, Humans, instrumentation/trends, Knowledge Bases, Models, National Institutes of Health (U S ), Neurosurgical Procedures, Organizational, Outcome Assessment (Health Care), Reproducibility of Results, Robotics, Software, Surgery, Treatment Outcome, United States}, doi = {10 1016/j neuroimage 2007 04 026}, url = {http://dx doi org/10 1016/j neuroimage 2007 04 026}, author = {Simon DiMaio and Tina Kapur and Kevin Cleary and Stephen Aylward and Peter Kazanzides and Kirby Vosburgh and Randy Ellis and James Duncan and Keyvan Farahani and Heinz Lemke and T. M. Peters and William Bill Lorensen and David Gobbi and John Haller and Laurence Larry Clarke and Stephen Pizer and Russell H. Taylor and Robert L. Galloway, Jr. and Gabor Fichtinger and Nobuhiko Hata and Kimberly Lawson and Clare M Tempany and Ron Kikinis and Ferenc Jolesz} } @article {Boctor2006b, title = {Ultrasound monitoring of tissue ablation via deformation model and shape priors}, journal = {Medical image computing and computer-assisted intervention (MICCAI)}, volume = {9}, number = {Pt 2}, year = {2006}, pages = {405{\textendash}412}, abstract = {

A rapid approach to monitor ablative therapy through optimizing shape, elasticity parameters is introduced Our motivating clinical application is targeting, intraoperative monitoring of hepatic tumor thermal ablation, but the method translates to the generic problem of encapsulated stiff masses (solid organs, tumors, ablated lesions, etc ) in ultrasound imaging The approach involves the integration of the following components: a biomechanical computational model of the tissue, a correlation approach to estimate/track tissue deformation, and an optimization method to solve the inverse problem, recover the shape parameters in the volume of interest Successful convergence, reliability studies were conducted on simulated data Then ex-vivo studies were performed on 18 ex-vivo bovine liver samples previously ablated under ultrasound monitoring in controlled laboratory environment While B-mode ultrasound does not clearly identify the development of necrotic lesions, the proposed technique can potentially segment the ablation zone The same framework can also yield both partial, full elasticity reconstruction

}, keywords = {Algorithms, Animals, Automated, Biological, Cattle, Computer-Assisted, Elasticity, Hepatectomy, Image Enhancement, Image Interpretation, Imaging, Interventional, Liver, Mechanical, methods, Models, Pattern Recognition, physiology/surgery/ultrasonography, Reproducibility of Results, Sensitivity, Specificity, Statistical, Stress, Subtraction Technique, Surgery, Three-Dimensional, Ultrasonography}, author = {Emad Boctor and Michelle de Oliveira and Michael Choti and Roger Ghanem and Russell H. Taylor and Gregory D. Hager and Gabor Fichtinger} } @article {Masamune2001, title = {System for robotically assisted percutaneous procedures with computed tomography guidance}, journal = {Journal of Computer Aided Surgery}, volume = {6}, number = {6}, year = {2001}, pages = {370{\textendash}383}, abstract = {

We present the prototype of an image-guided robotic system for accurate, consistent placement of percutaneous needles in soft-tissue targets under CT guidance inside the gantry of a CT scanner The couch-mounted system consists of a seven-degrees-of-freedom passive mounting arm, a remote center-of-motion robot, and a motorized needle-insertion device Single-image-based coregistration of the robot, image space is achieved by stereotactic localization using a miniature version of the BRW head frame built into the radiolucent needle driver The surgeon plans, controls the intervention in the scanner room on a desktop computer that receives DICOM images from the scanner The system does not need calibration, employs pure image-based registration, and does not utilize any vendor-specific hardware or software features In the open air, where there is no needle-tissue interaction, we systematically achieved an accuracy better than 1 mm in hitting targets at 5-8 cm from the fulcrum point In the phantom, the orientation accuracy was 0 6 degrees, and the distance between the needle tip, the target was 1 04 mm Experiments indicated that this robotic system is suitable for a variety of percutaneous clinical applications

}, keywords = {Biopsy, Computer-Assisted, Humans, Image Processing, Imaging, instrumentation, instrumentation/methods, Needle, Phantoms, Robotics, Surgery, Tomography, User-Computer Interface, X-Ray Computed}, doi = {10 1002/igs 10024}, url = {http://dx doi org/10 1002/igs 10024}, author = {Ken Masamune and Gabor Fichtinger and Alexandru Patriciu and Robert Susil and Russell H. Taylor and Louis Kavoussi and James Anderson and Ichiro Sakuma and Takeyoshi Dohi and Dan Stoianovici} }