@article {Fichtinger2008b, title = {Robotic assistance for ultrasound-guided prostate brachytherapy}, journal = {Medical Image Analysis}, volume = {12}, number = {5}, year = {2008}, month = {06/2008}, pages = {535}, publisher = {Queen{\textquoteright}s University; 25 Union Street;725 Goodwin Hall; Kingston; ON; Canada gabor@cs queensu ca}, abstract = {

We present a robotically assisted prostate brachytherapy system, test results in training phantoms, Phase-I clinical trials The system consists of a transrectal ultrasound (TRUS), a spatially co-registered robot,fully integrated with an FDA-approved commercial treatment planning system The salient feature of the system is a small parallel robot affixed to the mounting posts of the template The robot replaces the template interchangeably,using the same coordinate system Established clinical hardware,workflow, calibration remain intact In all phantom experiments,we recorded the first insertion attempt without adjustment All clinically relevant locations in the prostate were reached Non-parallel needle trajectories were achieved The pre-insertion transverse, rotational errors (measured with a Polaris optical tracker relative to the template{\textquoteright}s coordinate frame) were 0 25 mm (STD=0 17 mm), 0 75 degrees (STD=0 37 degrees) In phantoms,needle tip placement errors measured in TRUS were 1 04 mm (STD=0 50mm) A Phase-I clinical feasibility, safety trial has been successfully completed with the system We encountered needle tip positioning errors of a magnitude greater than 4mm in only 2 of 179 robotically guided needles,in contrast to manual template guidance where errors of this magnitude are much more common Further clinical trials are necessary to determine whether the apparent benefits of the robotic assistant will lead to improvements in clinical efficacy, outcomes

}, keywords = {Brachytherapy, Computer-Assisted, Equipment Design, Equipment Failure Analysis, Humans, instrumentation, instrumentation/methods, Interventional, Male, Prostatic Neoplasms, Radiotherapy, radiotherapy/ultrasonography, Robotics, Ultrasonography}, doi = {10 1016/j media 2008 06 002}, url = {http://dx doi org/10 1016/j media 2008 06 002}, author = {Gabor Fichtinger and Jonathan Fiene and Christopher Kennedy and Gernot Kronreif and Iulian Iordachita and Danny Y Song and E. Clif Burdette and Peter Kazanzides} } @article {Fichtinger2007c, title = {Robotic assistance for ultrasound guided prostate brachytherapy}, journal = {Medical Image Computing and Computer-Assisted Intervention (MICCAI)}, volume = {4791}, year = {2007}, pages = {119{\textendash}127}, publisher = {The Johns Hopkins University, Baltimore, Maryland, USA gabor@cs jhu edu}, abstract = {

We present a robotically assisted prostate brachytherapy system, test results in training phantoms The system consists of a transrectal ultrasound (TRUS), a spatially co-registered robot integrated with an FDA-approved commercial treatment planning system The salient feature of the system is a small parallel robot affixed to the mounting posts of the template The robot replaces the template interchangeably, uses the same coordinate system Established clinical hardware, workflow, calibration are left intact In these experiments, we recorded the first insertion attempt without adjustment All clinically relevant locations were reached Non-parallel needle trajectories were achieved The pre-insertion transverse, rotational errors (measured with Polaris optical tracker relative to the template{\textquoteright}s coordinate frame) were 0 25mm (STD = 0 17mm), 0 75 degrees (STD = 0 37 degrees) The needle tip placement errors measured in TRUS were 1 04mm (STD = 0 50mm) The system is in Phase-I clinical feasibility, safety trials, under Institutional Review Board approval

}, keywords = {Brachytherapy, Computer-Assisted, Equipment Design, Equipment Failure Analysis, Humans, instrumentation/methods, Interventional, Male, Prostatic Neoplasms, Radiotherapy, Radiotherapy Planning, radiotherapy/ultrasonography, Robotics, Ultrasonography}, author = {Gabor Fichtinger and Jonathan Fiene and Christopher Kennedy and Gernot Kronreif and Iulian Iordachita and Danny Y Song and E. Clif Burdette and Peter Kazanzides} } @article {Fichtinger2006, title = {Robotically assisted prostate brachytherapy with transrectal ultrasound guidance{\textendash}Phantom experiments}, journal = {Brachytherapy}, volume = {5}, number = {1}, year = {2006}, pages = {14{\textendash}26}, publisher = {Engineering Research Center, The Johns Hopkins University, Baltimore, MD 21218-2682, USA gabor@cs jhu edu}, abstract = {

PURPOSE: To report the preliminary experimental results obtained with a robot-assisted transrectal ultrasound (TRUS)-guided prostate brachytherapy system METHODS AND MATERIALS: The system consists of a TRUS unit, a spatially coregistered needle insertion robot, and an FDA-approved treatment planning, image-registered implant system The robot receives each entry/target coordinate pair of the implant plan, inserts a preloaded needle, and then the seeds are deposited The needles/sources are tracked in TRUS, thus allowing the plan to be updated as the procedure progresses RESULTS: The first insertion attempt was recorded for each needle, without adjustment All clinically relevant locations were reached in a prostate phantom Nonparallel, parallel needle trajectories were demonstrated Based on TRUS, the average transverse placement error was 2 mm (worst case 2 5 mm, 80\% less than 2 mm), and the average sagittal error was 2 5 mm (worst case 5 0 mm, 70\% less than 2 5 mm) CONCLUSIONS: The concept, technical viability of robot-assisted brachytherapy were demonstrated in phantoms The kinematically decoupled robotic assistant device is inherently safe Overall performance was promising, but further optimization is necessary to prove the possibility of improved dosimetry

}, keywords = {Anatomic, Brachytherapy, Endosonography, Equipment Design, Humans, instrumentation, Male, methods, Models, Prostatic Neoplasms, radiotherapy/ultrasonography, Rectum, Robotics}, doi = {10 1016/j brachy 2005 10 003}, url = {http://dx doi org/10 1016/j brachy 2005 10 003}, author = {Gabor Fichtinger and E. Clif Burdette and Attila Tanacs and Alexandru Patriciu and Dumitru Mazilu and Whitcomb, Louis L. and Dan Stoianovici} }